Interactive education system for teaching patient care

ABSTRACT

An interactive education system for teaching patient care to a user is described. The system comprises a patient simulator; a virtual instrument for use with the patient simulator in performing patient care activities; means for sensing an interaction between the virtual instrument and the simulator, and means for providing feedback to the user regarding the interaction between the virtual instrument and the simulator.

CROSS-REFERENCE TO RELATED APPLICATION

[0001] This application is a continuation of U.S. Ser. No. 09/684,030, filed on Oct. 6, 2000, which is a continuation-in-part of U.S. Ser. No. 09/640,700, filed Aug. 17, 2000, which is a continuation-in-part of U.S. Ser. No. 09/560,949, filed Apr. 28, 2000, which is a continuation-in-part of U.S. Ser. No. 09/199,599, filed Nov. 25, 1998, which is a continuation of U.S. Ser. No. 08/643,435, now U.S. Pat. No. 5,853,292, filed May 8, 1996. The entire disclosures of the foregoing applications are hereby incorporated by reference.

BACKGROUND

[0002] The present embodiment relates generally to an interactive education system for teaching patient care, and more particularly to such a system having virtual instruments for use with a child birthing patient simulator in conducting patient care activity.

[0003] While it is desirable to train students in patient care protocols before allowing contact with real patients, textbooks and flash cards lack the important benefit to students attained from “hands-on” practice. Thus, patient care education has often been taught using medical instruments to perform patient care activity on a simulator, such as a manikin. However, one disadvantage of such a system is that medical instruments are often prohibitively expensive, and consequently, many users must settle for using a smaller variety of instruments, even at the cost of a less comprehensive educational experience. One solution to the foregoing problem is using a set of relatively inexpensive, simulated medical instruments (“virtual” instruments), as taught in U.S. Pat. No. 5,853,292, the entire disclosure of which is hereby incorporated by reference.

[0004] Another problem in patient care education is that the patient simulators used for teaching a user are generally passive. For example, in a child birthing simulation, a user must position the simulated fetus in a simulated maternal pelvis, move it down the birth canal, birth the fetus's head, rotate the fetus approximately ninety degrees to birth the shoulders, and finally, pull out the fetus, now referred to as a neonate. While replicating the sequence of events in a real delivery, the lack of verisimilitude resulting from physical manipulation of the fetus by the user undermines an appreciation for the difficulties of providing patient care. In a real delivery, the fetus is inaccessible, and most activity is obscured from view, and thus prior systems fail to address the most challenging conditions of providing patient care during child birthing. Moreover, prior systems fail to simulate cervical dilation as the fetus moves down the birth canal, thus failing to allow a student to assess the stage of delivery or construct a chart of cervical dilation versus time to assess the progress of delivery (“Partograph”).

[0005] Therefore, what is needed is a system for an interactive education system for use in conducting patient care training sessions using relatively inexpensive virtual instruments in cooperation with a more realistic simulated patient, thereby enabling a user to learn comprehensive multiple and interrelated patient care skills.

SUMMARY

[0006] The present embodiment provides an interactive education system for teaching patient care to a user. The system includes a maternal simulator, a fetal simulator designed to be used both in conjunction with the maternal simulator and separate from the maternal simulator, and neonatal simulator designed to replace the fetal simulator in post-birth simulations.

BRIEF DESCRIPTION OF THE DRAWINGS

[0007]FIG. 1a is a schematic view of an illustrative embodiment of an interactive education system.

[0008]FIG. 1b is a schematic view of an interactive education system according to another embodiment.

[0009]FIG. 2 is a schematic view of the interaction between a set of virtual instruments and a patient simulator.

[0010]FIG. 3a is a perspective view with a cutaway of a virtual instrument.

[0011]FIG. 3b is a perspective view with a cutaway of a sensor.

[0012]FIG. 4 is a perspective view of an illustrative embodiment of a patient simulator.

[0013]FIG. 5a is a perspective view of the patient simulator of FIG. 4 with an attached cover.

[0014]FIG. 5b is a top plan view of a control box.

[0015]FIG. 6 is a perspective view of the torso of the patient simulator of FIG. 4.

[0016]FIG. 7 is a perspective view of FIG. 6 with the fetal portion of the patient simulator removed.

[0017]FIG. 8 is a perspective view of a distensible cervix of the patient simulator.

[0018]FIG. 9 is a perspective view of the exterior of the patient simulator.

[0019]FIG. 10 is a perspective view of a neonatal embodiment of a patient simulator.

[0020]FIG. 11 is a schematic view of an illustrative use of the present system.

[0021] FIGS. 12-16 are screen display views generated by a program according to one embodiment of the present system.

DETAILED DESCRIPTION

[0022] Referring to FIG. 1a, the reference numeral 10 refers, in general, to an interactive education system for teaching patient care protocols to a user. The system 10 comprises a set of virtual instruments 12 used to simulate medical instruments, and a simulator 14 used to simulate at least one patient for receiving patient care activity from the user. The virtual instruments 12 are tangible objects, and look, feel, and operate like real medical devices in conjunction with the simulator 14, which is understood to encompass a variety of forms, including a fully articulating and adult-sized manikin, as well as a fetus, a neonate, a child, a youth, or portion of a manikin, such as the arm, torso, head, or pelvic region.

[0023] Patient care activity received by the simulator 14 from the user, or users, is sensed in a manner to be described, and in response to the activity, the system 10 provides feedback to the user. It is understood that feedback may comprise any audio, visual, or tactile response. A computer 15 having a program 15 a is optionally connected to the system 10, for reasons to be described.

[0024] Referring to FIG. 1b, a system 10′ comprises the computer 15 and the program 15 a, wherein a software-generated set of virtual instruments 12′ and a software-generated simulator 14′ is provided. Thus, the patient care activity performed by the user comprises manipulating an icon relating to a selected software-generated virtual instrument 12′ to provide patient care to the software-generated simulator 14′. In this embodiment, the program 15 a uses conventional means, such as clicking a mouse or voice-activated software, to monitor activity by the user, and provides feedback in response, as will be described.

[0025] Returning to FIG. 1a, the system 10 further comprises a communications interface module (“CIM”) 16, which receives operating power from a conventional power source 18, and contains a microcontroller (“PIC”) 20. Microcontrollers are available from many vendors, such as Microchip Technology, Inc. (Chandler, Ariz.), and are then customized. As will be described, the PIC 20 receives input signals from the user's activity, and is programmed to respond in a certain manner to provide feedback to the user. For example, to provide audio feedback, the CIM 16 additionally includes an audio chip 22 which is responsive to the PIC 20 for causing a speaker 24 to produce realistic patient sounds, for example, heart, lung, blood pressure (Korotkoff), intestinal, fetal, and the like. A control 26 is included in the CIM 16 for adjusting the volume of the speaker 24.

[0026] Alternatively, depending on the complexity of the desired feedback, the CIM 16 may be connected to the computer 15 and program 15 a. In one example of feedback, the program 15 a could be used to provide a vast library, for example, of ultrasound profiles, or fetal distress monitor traces. Feedback could also be of body sounds, generated by the program 15 a, and played through speakers of the computer.

[0027] The CIM 16 has a plurality of ports, collectively 28, for receiving input signals occasioned by interaction between the virtual instruments 12 and sensors 30 disposed on the simulator 14, resulting from the user's patient care activity. It is understood that there may be more than one PIC 20, and more than one CIM 16, to manage the input signals thus created.

[0028] The virtual instruments 12 comprise patient care devices, for example, as shown in FIG. 2, at least one IV needle, an endotracheal (ET) tube, an electrocardiogram (ECG or EKG) monitor, a blood pressure (BP) cuff, a pulse oximeter cuff, a temporary external pacer, an automatic external defibrillator (AED), a manual defibrillator, an ultrasound wand, a virtual stethoscope, a thermometer, and a fetal distress monitor, respectively 12 a-l. Such virtual instruments look and operate like real medical devices. Of course, other virtual instruments are contemplated, as is the use of relatively inexpensive medical devices, such as a conventional stethoscope, a vacuum extractor, catheters, trays, IV stands, and the like.

[0029] Referring to FIG. 2, the IV needle 12 a has a selectable group of specific drugs and dosages, and in one embodiment is part of a medication tray with an assortment of labeled syringes for dispensing the drugs to the simulator 14, with the effects of administration controlled by the program 15 a. The ET tube 12 b is used in simulated patient airway management, and placed in a tracheal airway of the simulator 14. The EKG monitor 12 c comprises a 3, 5, or 12 lead system, including a real-time trace monitor and R-wave sonic markers, and a plurality of color-coded patches for attachment to a torso of the simulator 14. The BP cuff 12 d attaches to the simulator 14, for example, around an arm. The pulse oximeter finger cuff 12 e attaches to the simulator 14, for example, around a finger. The temporary external pacer 12 f has a plurality of anterior and posterior pacer pads for attachment to the torso of the simulator 14. The pacer 12 f has controls for pacer rate and current, and exhibits rhythm pacing, cap time, and loss of cap time, all of which is controlled by the program 15 a. The automatic external defibrillator (AED) 12 g has a plurality of apex and sternum AED pads for attachment to the torso of the simulator 14. Upon selecting a software-generated shock button produced by the program 15 a, the system 10 simulates defibrillation shock, with the resultant conditions controlled by the program 15 a. The manual defibrillator 12 h has a plurality of apex and sternum defibrillator paddles for contacting the torso of the simulator 14. Upon selecting a software-generated shock button, or alternatively by using a dual shock buttons associated with manual defibrillator 12 h, the system 10 simulates defibrillation shock, with the resultant conditions controlled by the program 15 a.

[0030] Still referring to FIG. 2, the ultrasound wand 12 i interacts with the simulator 14, such that when the wand 30 i is brought within a predetermined proximity of a predetermined anatomical area of the simulator, the CIM 16 detects the interaction and the program 15 a supplies an ultrasound profile taken from a library of ultrasound images and or sounds. The program 15 a may select between normal and abnormal profiles, requiring the user to interpret the profile and respond accordingly. The virtual stethoscope 12 j interacts with the simulator 14, such that when the stethoscope 12 j is brought within a predetermined proximity of a predetermined anatomical area of the simulator, the CIM 16 detects the interaction and feedback is supplied to the user, as will be explained below, with FIGS. 3a-b. The thermometer 12 k interacts with the simulator 14, such that when the thermometer 12 k is brought within a predetermined proximity of a predetermined anatomical area of the simulator, the CIM detects the interaction and the program 15 a supplies a temperature reading. The fetal distress monitor 12 l (tocodynomometer) attaches to a portion of the simulator 14, and upon attachment, the program 15 a supplies a heart rate reading for a simulated fetus.

[0031] Each instrument has a corresponding sensor 30 a-l, as indicated by lines, collectively 36. Unless otherwise indicated, the lines 36 are schematic, and merely illustrate that the virtual instruments 12 and the sensors 30 are functionally connected to each other for providing an interaction created by the user's patient care activity, the interaction being reported as an input signal to the CIM 16. It is understood that the sharing of such physical lines among instruments 12, or sensors 30, is contemplated as well.

[0032] Interaction between the virtual instruments 12 and the sensors 30 may be electrical, optical, pressure differential, tactile, temperature-controlled, or wireless. Generally speaking, an electrical interaction (which would also provide the input signal) could be created via a virtual instrument 12 having one node and a sensor 30 with another node, both of which are physically connected to the CIM 16, or by a virtual instrument with two nodes and a sensor formed of conductive material, or vice versa, only one of which may be physically connected to the CIM 16. For example, the IV needle 12 a corresponds with a portion of the simulator 14 capable of accepting medications, such as the antecubital region of an arm, which may have a sensor 30 a comprising an insulator sandwiched between two layers of conductive material having an appropriate thickness and weave density for permitting the needle 12 a to pass through the cloth at a low acute angle (e.g., 20°). The conductive layers of the sensor 30 a are electrically coupled to the CIM 16 via line 36 a′, such that when the needle 12 a is correctly passed through the two conductive layers, simulating cannulation of a vein of the simulator 14, a circuit is completed between the layers and sensed by the CIM 16.

[0033] In another example of a method of sensing interaction, the ET tube 12 b is used in simulated patient airway management, the simulator 14 having a head, eyes, a nose, a mouth, and a realistic airway capable of accepting conventional airway adjuncts, with the airway configuration adjustable to display a large tongue, an obstructed pharynx, or closed vocal cords, to increase the difficulty of the patient care activity. In order to confirm proper placement in the tracheal airway of the simulator 14, an optical sensor 30 b is mounted in the wall of the trachea of the simulator 14 and connected to the CIM 16 via line 36 b′. Correct placement of the ET tube 12 b in the trachea is confirmed when the tip of the ET tube interrupts the beam of the optical sensor 30 b. The sensor 30 b may also be used to determine whether a fluid has passed.

[0034] The virtual stethoscope 12 j provides an example of a wireless method of sensing interaction. At least one sensor 30 j is placed at an anatomical location on the simulator 14 where specific heart, lung (including airway), Korotkoff, fetal, or other sounds are normally heard. The sensor 30 j provides at least one signal which is identified by the stethoscope 12 j, thereby directing an integrated sound circuit to play a sound to the user appropriate for the anatomical location of the sensor on the simulator 14. It is understood that the sound circuit has a stored library of body sounds corresponding to the location of the selected sensor 30 j, and that the sensor 30 j is illustrative of any number of similar sensors.

[0035] Referring to FIG. 3a, in some respects, the appearance of the stethoscope 12 j resembles a standard stethoscope, having earpieces 50 a-b for hearing sounds, and being connected to extenders 51 a-b, which are joined to a bifurcated ear tube 52. Similarly, the stethoscope further comprises a bell tube 54, and a bell 56, preferably made of nonferrous material. However, unlike conventional stethoscopes, an electronic control box 58 is disposed between the ear tube 52 and the bell tube 54. The control box 58 is understood to be an appropriately developed CIM 16, physically integrated into the virtual instrument 12 j, thus simplifying the system 10. A jack 64 is provided on the control box 58 for output to an external speaker (not depicted), so that other users may hear the sounds heard in the earpieces 50 a-b. This not only increases the number of users who benefit from the patient care activity, but allows an instructor to test the user's ability, and correct the user's technique if required. The control box 58 retains a small power source 66, such as a battery, an acquisition circuit 68 and a sound circuit 70 (see copending U.S. application Ser. No. 09/640,700, filed Aug. 17, 2000, for circuit diagrams) for directing a small speaker 72, such as is available from ADDAX Sound Company (Northbrook, Ill.), to play a predetermined sound. The speaker 72 is disposed in the earpiece 50 a, and connected to the control box 58 via a wire 72 a, allowing the user to hear the sounds produced by the sound circuit 70. It is understood that a second, substantially identical speaker may be disposed in the opposite earpiece 50 b, and also connected to the control box 58. In an alternative embodiment, the speaker 72 may be disposed in the control box 58, and sounds transmitted via conventional ear tubes to the ear pieces. The sound circuit 70 is also connected to the jack 64 for allowing connection to an external speaker for the above-described reasons.

[0036] A switch 74, having a number of positions, is disposed on the control box 58 for switching between groups of sounds, for example exemplary normal and abnormal sounds that may be those heard in an adult, neonate, or fetus. An RF (radio frequency) signal acquisition coil 76, such as is available from M. C. Davis Co. (Arizona City, Ariz.), is disposed in the interior of the bell 56 for transmitting and acquiring RF signals, as will be explained. The acquisition coil 76 is a copper coil and circuitry having an associated wire 76 a, which is attached to the electronic control box 58. A polymeric disc 78 is disposed between the acquisition coil 76 and the bell 56 to decrease noise from the bell.

[0037] Referring to FIG. 3b, the sensor 30 j is disposed beneath the skin 14 b of the simulator 14 to avoid visual detection by the user. Likewise, it is advantageous that the sensor 30 j have a minimal thickness to prevent intentional or accidental detection, as some anatomical locations, for example, intercostal spaces, must be palpated in order to be located. In an alternative embodiment, the sensors 30 j may be affixed to an overlay (not depicted) substantially similar to the skin 14 b, thus allowing the overlay to be placed over other simulators and models of patients, thereby converting those devices to allow them to be used with the stethoscope 12 j.

[0038] The sensor 30 j comprises an RF ID tag 80, such as is available from Microchip Technology, Inc. (Chandler, Ariz.) (Part No. MCRF200-I/3C00A), which may be programmed using “Developer's Tools” also sold by Microchip Technology, Inc. to engender a unique signal that serves to identify the particular sensor 30 j. A coil 82, such as is available from M. C. Davis Co. (Arizona City, Ariz.), is operably connected to the tag 80. The tag 80 and coil 82 are potted in RTV potting material 84, or silicon rubber, such as is available from M. C. Davis Co. (Arizona City, Ariz.), to prevent damage. Once potted, the tag 80 and coil 82 collectively form a COB module 86 which emits a signal comprising a unique train of frequencies when interrogated.

[0039] In operation, the COB module 86 may actively broadcast the frequencies, but preferably the COB module is passive, that is, only activated when interrogated by the acquisition coil 76 in the stethoscope bell 56. In this preferred embodiment, the acquisition coil 76 delivers a carrier signal, such as a 125 kHz excitation frequency, which is received by the COB module 86 when the bell 56 is brought within a predetermined proximity, or acquisition distance, of the COB module. The acquisition distance of the bell 56, and therefore the acquisition coil 76, to the COB module 86 is determined by the strength to noise (S/N) ratio of the carrier signal. Thus, adjustment of the S/N ratio of the carrier signal provides a means for controlling the precision with which the user must place the stethoscope bell 56 in relation to the anatomical location of the sensor 30 j, and therefore the COB module 86. Precise placement of the bell 56 on the simulator 14 by the user is rewarded with feedback, in the form of an appropriate body sound. Normally, the S/N ratio is set to require that the bell 56 be brought within approximately one-half to two centimeters of the COB module 86 of the sensor 30 j.

[0040] In response to receiving a sufficiently strong carrier signal, the COB module 86 emits a train of two identifying frequencies for use in a process conventionally known as frequency shift keying (FSK), although other keying methods could be used. The acquisition coil 76 in the stethoscope bell 56 receives the emitted frequencies and relays the signal to the acquisition circuit 68, which determines the identity of the sensor 30 j. As the anatomical position of each sensor 30 j is known to the programmer, a selection of appropriate body sounds associated with each sensor is provided, and accessible to the sound circuit 70. Thus, by identifying the sensor 30 j, the acquisition circuit 68 directs the sound circuit 70 to play an appropriate body sound for the anatomical position of the COB module 86, which is heard by the user through the speaker 72 disposed in the earpiece 50 a. It can be appreciated that to expose the user to a greater selection of sounds, more sensors 30 j could be added to the simulator 14, or each sensor could correspond to more than one sound. As depicted, the switch 74 has five different positions, and includes means for switching the sound circuit 70 between five different groups of sounds. Thus, it is understood that the number of switch positions corresponds to the number of sounds that can be produced by a single sensor, i.e., with thirteen sensors and five switch positions, the user could listen to up to sixty-five location-appropriate sounds, including examples of normal and abnormal sounds.

[0041] It can be appreciated that the above-described acquisition coil and COB module may be adapted to be used with the respective leads, paddles, or probes (“connectors”) of the ECG monitor 12 c, the temporary external pacer 12 f, the automatic external defibrillator (AED) 12 g, the manual defibrillator 12 h, the ultrasound wand 12 i, and the fetal distress monitor 12 l. If desired, the connectors may be equipped with adhesive to temporarily hold them in place on the patient simulator. The interaction between the instruments' connectors and the sensors 30, as sensed by the CIM 16, confirms proper placement. The hidden location of the sensors 30 beneath the skin of the patient simulator further challenges a user's patient care skills, as well as more closely mimicking a real patient.

[0042] It is understood that the simulator 14 is designed to represent a patient and receive treatment, and as such the simulator 14 could take a variety of forms, including a fully articulating and adult-sized obstetrics simulator, a curled fetus, an articulating fetus, multiple fetuses, or a neonate, as well as a portion of simulated patient, for example, the torso and pelvic region.

[0043] Referring to FIGS. 4 and 5a, in an illustrative embodiment, the simulator 14 comprises a child birthing maternal simulator 300 and a removable associated fetal simulator 302. The maternal simulator 300 has a head 304, with hair 306, eyes 308 a-b, a nose 310, and a mouth 312. The head assembly contains a realistic airway (not depicted) capable of accepting conventional airway adjuncts. Sensors, generally denoted 30 (FIG. 1a), may be disposed on the skin of the maternal simulator (shown as stippled) and/or beneath the skin (shown in phantom). It is understood that in one embodiment of the maternal simulator (not depicted), no sensors are associated with the simulator. Lines 36 protrude from the torso 316 for providing electrical, pneumatic, or fluid connections, as well as for connecting the sensors 30 to the CIM 16, if necessary.

[0044] A pair of arms 318 a-b are connected to the torso 316. At least one arm contains an IV receptacle (not depicted) capable of accepting medications, and sensors 30 a may be placed within the receptacle to ascertain whether an IV has been started. Similarly, the arm may contain a sensor 30 d for auscultation of Korotkoff sounds, as well as means for measurement of blood pressure. A pelvic region 320 of the torso 316 receives a pair of legs 322 a-b.

[0045] Referring to FIG. 5a, a cover 324 may be attached to the torso 316 via a plurality of snaps 324 a, although other reversible fastening means, such as hook and loop closures may be used. The cover 324 retains sensors 30, for cooperating with the ultrasound wand 12 i, fetal distress monitor 12 l, and the stethoscope 12 j, or alternatively at least one small speaker, to allow simulation of fetal heart sounds which may be detected by the stethoscope 12 j or a conventional stethoscope, respectively. In one embodiment, the cover 324 surrounds an open cell foam (not depicted) connected to means for producing a vacuum. Activation of the vacuum shrinks the foam, making it feel harder, which simulates uterine contractions by the maternal simulator 300. Alternatively, the cover 324 may retain an air bladder and associated line (not depicted) for pressurizing the cover, thus making it feel harder. It is understood that different levels of hardness may be produced to simulate different levels of contraction strength, for example, mild, moderate, and strong contractions. If connected to the CIM 16 and program 15 a, the contractions could be spaced at regular intervals, and associated data for maternal intrauterine pressure may be displayed by the program, as will be discussed with FIG. 14.

[0046] Returning to FIG. 4, the fetal simulator 302, has an umbilical cord 302 a and placenta 302 b, and is depicted as resting upon a removable stage 325 disposed inside the maternal simulator. The removable stage 325 has a bladder (not shown), a line 325 a, and a bulb 325 b. When the bulb 325 b is used to pump air into the bladder, the stage 325, and hence the fetal simulator 302, is raised relatively upwards. When covered with the cover 324 (FIG. 5a), raising of the stage 325 allows a user to palpate the fetal simulator 302 through the cover to assess position, as well as to perform Leopold maneuvers.

[0047] A birthing device 326 is disposed inside the torso 316, as will be described. The cover 324 is designed to obscure the fetal simulator 302 of the simulator and the birthing device 326 from view, thus more accurately simulating the child birthing process, and challenging the user's diagnostic abilities. With the stage 325 removed, the birthing device 326 may be operated via a manual crank (not shown), or by a small motor 326 a connected via a line 326 b to controlling means for turning the motor on or off, as well as determining operational speed.

[0048] In a first embodiment, software of the program 15 a controls the birthing device 326, as will be discussed in conjunction with FIG. 14, below. In an alternative embodiment, the controlling means is a control box 328, and a line 330 which connects the control box 328 to the CIM 16. Referring to FIG. 5b, the control box 328 has controls 328 a-d for respectively turning the simulator 14 on and off, pausing and resuming child birthing, determining the speed of the delivery rate, and setting the fetal heart rate.

[0049] Referring to FIGS. 6 and 7, the torso 316 of the maternal simulator 302 is shown with the cover 324 removed to expose the fetal simulator 302. The fetal simulator 302 is disposed in a cavity 333 of the maternal simulator 300, and has a head 334, an attached torso 336, with a pair of arms 338 a-b and legs 340 a-b attached to the torso. The head 334 is soft to allow for vacuum extraction, and has a mouth and nose which may be suctioned by the user. The umbilical cord and placenta 302 a-b (FIG. 4) are removed to simplify the illustration, but it is understood that the placenta 302 b (FIG. 4) could be disposed in any number of common orientations, such as normal fundal, low placement, or placenta previa, and attached to the cavity 333 with conventional removable fasteners. Likewise, the umbilical cord 302 a (FIG. 4) could be presented to replicate various complications, and may house connecting lines to the fetal simulator 302 to allow an umbilical pulse to be felt by the user, or to convey electricity to the fetal simulator 302, if necessary.

[0050] A receiver 342 is disposed on the fetal simulator 302 to allow the birthing device 326 to retain the fetal simulator. Other receivers, similar to the receiver 342, are contemplated on different portions of the fetal simulator 302, such as to simulate a breech birth, and as the fetal simulator 302 articulates, a variety of breech deliveries, such as full, frank, and footling may be simulated.

[0051] The birthing device 326 has a projection 344 of a ram 346 which cooperates with the receiver 342 of the fetal simulator 302 to retain the fetal simulator. In the depicted embodiment, the ram 346 is driven by a drive system, including a small electric motor, gears, electronic logic to permit resetting, means to determine the position of the ram, and a forward and reverse function. The ram 346 proceeds down a set of tracks 347 a-b, thereby translating the fetal simulator 302 out of the maternal simulator 300.

[0052] The projection 344 of the ram 346 is rotatable, the birthing device 326 thereby producing both rotational and translational movement of fetal simulator 302, to simulate a realistic child birthing scenario, wherein the fetus makes a turn to bring it to a normal nose down position of crowning, and it makes another turn after crowning to allow its shoulders to better pass through the birth canal.

[0053] In one embodiment, levers 346 a-b of the ram 346, being operably connected to the projection 344, engage cams 348 a-b, respectively, to produce rotation. As the ram 346 proceeds down the tracks 347 a-b, the levers 346 a-b of the ram engage the fixed cams 348 a-b in turn, causing the respective lever to move. Movement of the lever rotates the projection 344. Eventually, the respective lever is moved to a point where the lever clears the respective cam. It can be appreciated that the cams 348 a-b may be located at places along the tracks 347 a-b where rotation is desired, the tracks simulating the birth canal. Thus, internal rotation of the fetus is produced by the lever 346 a engaging the cam 348 a, and external rotation of the fetus is produced by the lever 346 b engaging the cam 348 b. Alternatively, the program 15 a allows for adjustment of the rotation of the projection 344 from zero to one hundred and eighty degrees, as will be discussed with reference to FIG. 14, below. In either embodiment, the fetus 302 passes through a distensible cervix 350, as will be described.

[0054] Referring now to FIGS. 8 and 9, the distensible cervix 350 comprises a ring 352 having attached flaps 353 a-b for maintaining the cervix's position in the cavity 333. As such, the flaps 353 a-b may have attached snaps, hook and loop closures, or other reversible fastening means. A wall 354 is connected to the ring 352, and is preferably of an elastic material, such as Lycra®. A gathering 356 of the wall material defines a port 358. The gathering 356 may have an associated elastomeric element disposed interiorly to enhance the elasticity of the port 358. Alternatively, the wall 354 itself may provide sufficient elasticity.

[0055] The port 358 expands from about two to ten centimeters in diameter as the fetal simulator 302 is pushed through the port, and because of the shape of the fetal simulator's head 334, and the elasticity of the wall 354, dilation is automatically simulated coincident to fetal descent. The user may then practice measuring cervical dilation and plot labor progress as a Partograph. The elasticity of the wall 354 may be adjusted, for example by using thicker or thinner wall material, to produce a cervix having faster or slower dilation than normal, respectively. The cervix 350 is disposed concentric to the pelvic area 320, which has a pubic bone 360, as well as several cover snaps 324 a.

[0056] The fetal simulator 302 moves through the cervix 350 and out of the cavity 333 past vulva 362. The vulva 362 are made of a flexible material so that the user may manipulate the vulva, or perform an episotomy to birth the head 334. It is understood that the vulva 362 may comprise a portion of an insert (not depicted) including features such as a urinary tract and rectum, which could be replaceable with other genital inserts for displaying various patient conditions. After delivery, the user may practice postpartum exercises, such as massaging a uterus insert (not depicted) back to a desirable size, removing retained placenta parts (not depicted), or repairing the cervix 350 or vulva 362.

[0057] In one embodiment, the torso 316 contains a simulated heart, lungs, and ribs. The heart (not depicted) beats by the action of a pulsatile flow which is controlled by the program 15 a in response to the condition of the patient and upon therapeutic interventions. Palpable pulses may be found at carotid, brachial, radial, femoral, and pedis dorsis locations. Specific pulse locations become non-palpable as the systolic pressure falls, and the absence or presence of a pulse will depend upon the simulated blood pressure. Heart sounds are heard at appropriate locations through the use of the stethoscope 12 j. The heart beat is synchronized with the Virtual EKGs, which are determined by the program 15 a. Application of the stethoscope 12 j to a point below the BP cuff 30 d (FIG. 2) will cause the appropriate Korotkoff sounds to be heard.

[0058] The maternal simulator 300 displays a combination of ventilation means, and lung and airway sounds are heard at appropriate locations using the stethoscope 12 j. The simulator 300 breathes spontaneously in a manner that would achieve targeted arterial blood gases for a given situation, including response to interventions such as ventilation and administration of drugs, and demonstrates the amount of chest rise relating to the tidal volume and physiologic states. Normal gas exchange lung dynamics are virtual and are controlled by the program 15 a, which may also determine tidal volumes (TV), functional residual capacity (FRC), and expired carbon dioxide (CO₂). Airway resistance, lung and chest wall compliance are also controlled by the program 15 a.

[0059] The heart and lungs are connected to pressure transducers confirming airway ventilation and cardiac compression. For example, an air line may be mounted in tracheal wall or lungs of the simulator 300 and connected to a sensor circuit connected to the CIM 16 so that when cardiopulmonary resuscitation (CPR) ventilation is performed on the simulator, the CIM 16 monitors the timing and magnitude of the pressure and volume of the ventilation procedure, via the air line and the sensor. Similarly, a compression bladder may be embedded within the heart or chest cavity of the simulator 300 for sensing and confirming proper timing and magnitude of a CPR chest compression procedure, when connected by an air line to a compression sensor circuit attached to the CIM 16. It can be appreciated that compression and ventilation data is acquired from pressure waves sensed by the CIM 16 through the lines 36. The blood pressure, heart rate, and oxygen saturation is virtually measured by the BP cuff 30 d (FIG. 2) and the Pulse Ox cuff 30 e (FIG. 2), although the data displayed is generated by the program 15 a.

[0060] Referring to FIG. 10, a neonate simulator 302′ may be used to replace the fetal simulator 302 (FIG. 8) to allow practice of neonatal resuscitation according to the program 15 a. The neonate 302′ has a head 370, with hair 372, eyes 374 a-b, a nose 376, and a mouth 378. The head assembly contains a realistic airway (not depicted) capable of accepting conventional airway adjuncts and a sensor for determining whether an airway adjunct has been placed, or whether a fluid has passed. The head 370 is connected via a neck 380 to a torso 382.

[0061] Sensors, generally denoted 30 (FIG. 1a), may be disposed on the skin of the neonate simulator (shown as stippled) and/or beneath the skin (shown in phantom). Lines 36″ protrude from the torso 382 for providing electrical, pneumatic, or fluid connection, as well as for connecting sensors (not depicted) to the CIM 16. The torso 382 has an umbilical site 384, which provides a site for catheterization, and a simulated heart, lungs, and ribs for performing CPR. The heart and lungs are connected to pressure transducers as described above for the maternal simulator 300 for confirming airway ventilation and cardiac compression. The neonate simulator 302′ exhibits many of the same features as the maternal simulator 300 (FIG. 6), including heart rate, pulse, oxygenation, and a variety of body sounds which can be detected using the stethoscope 12 j (FIG. 2) or a conventional stethoscope. A pair of arms 386 a-b, and a pair of legs 388 a-b, are also connected to the torso 3382.

[0062] In one embodiment, the hands and feet as well as the face and upper torso change color based upon proper oxygenation or an oxygen deficit. As oxygenation decreases, the extremities (peripheral cyanosis) change color first, followed by the face and upper torso (central cyanosis). Such change is reversible as oxygenation is improved.

[0063] In a preferred embodiment, coloration is achieved using blue thermochromatic dye (such as Reversatherm Blue Type F, available from Keystone, Chicago, Ill.), approximately 3 grams dissolved in 10 grams of clear vinyl paint thinner, and dispersed into 300 grams of clear vinyl paint. The mixture is applied to the hands, feet, chest, and face. At room temperature, the neonate is blue. Resistance heaters (such as available from Minco Products, Minneapolis, Minn.) are connected in parallel, and placed under the skin to provide 5-15 watts/in², or heat energy sufficient to raise the surface temperature of the skin to about 115°, causing the bluish color to disappear. Power for the heater is supplied through the CIM 16. The peripheral and central heaters may be separately controlled to allow peripheral cyanosis without central cyanosis. Heat sinks may also be disposed with the heaters to allow faster cooling, and hence, faster changes in coloration.

[0064] In one embodiment, the thermochromatic system is logically linked to the program 15 a, for example, an instructor defines the condition of the neonate. Afterwards, coloration is responsive to CPR quality being performed by a user, either improving, worsening, or remaining the same. The program 15 a also provides for an override if coloration changes are not desired. Coloration may alternatively be simulated by having applied a conventional photochrome to the simulator, such that upon exposure to an associated adjustable UV light, the simulator appears to turn blue.

[0065] Referring now to FIG. 11, a child birthing system 500 illustrates the use of the foregoing embodiments. The simulator 14, for example, the maternal simulator 300 and fetus 302 are placed on a table 502. Students, W, X, Y, and Z, take places around the table, for example, W controls medication, Y controls virtual instruments 12, X controls anesthesia, and Z controls obstetrics. The child birthing device 326, as discussed above, may be driven via a manual crank or by a small motor 326 a connected to either a control box 328, or the program 15 a of the computer 15 may optionally (shown in phantom) control the birthing device 326. Whichever controlling means are used, the distensible cervix accurately reflects progress of the fetal simulator down the birth canal. Eventually, as described above, the fetal simulator is birthed.

[0066] Once the fetal simulator is birthed, a team W′, X′, and Y′ (which are understood to be the same students W, X, and Y, or others depending on class size) moves along path 1 to practice neonatal care on a table 502′. At least one team, denoted by the absence of Z, must remain behind with the maternal simulator for monitoring and potential stabilization. The fetal simulator is switched with a neonatal simulator 14′, for example, neonatal simulator 302′ (FIG. 10). If connected to the computer, the program 15 a may be used to simulate the need for neonatal resuscitation, and CPR and other emergency care protocols may be performed. The program 15 a monitors the care received by the simulator via the CIM 16 and virtual instruments 12, and compares the care to accepted standards.

[0067] Meanwhile, the program 15 a of the computer 15 may be used to simulate the need for maternal resuscitation. If so, a team moves along path 2 to practice maternal care on a table 502″. Students, W″, X″, Y″, and Z can work on the maternal simulator 14″, for example maternal simulator 300 with the fetal simulator removed. CPR and other emergency care may be given, and the program 15 a monitors the care received by the simulator via the CIM 16 and virtual instruments 12.

[0068] Referring now to FIG. 12, an introductory screen display 400 of the program 15 a is presented on the computer 15 for teaching patient care protocols to a user. The display 400 includes several decorative features: a title box 402, a fetal heart rate box 404, a maternal intrauterine pressure box 405, a vital signs box 406, and an ultrasound video box 407. The display 400 also contains a teaching box 408, a testing box 410, and a virtual instruments box 412. As will be described, in some modules, the program 15 a compares information pertaining to the user's activity with predetermined standards.

[0069] The screen 400 also displays a group of selectable patient care modules 414 a-p provided by the program 15 a, which furnish information on medical topics and associated concepts. Each module has a single topic, and represents an interactive patient care training session for the user. The modules 414 a-g are disposed in the teaching box 408, and give an overview of relevant physiology, pregnancy, complications, labor and birth, postpartum, and maternal and neonatal resuscitation protocols. The modules 414 h-j are disposed in the testing box 410, and give an opportunity to test a user in maternal and neonatal resuscitation protocols, as well as instructor defined protocols (Codemaker). An exit button 415 for exiting the program 15 a is also disposed in the testing box 410. The modules 414 k-p are disposed in the virtual instruments tutor box 412, and give a user a tutorial on use of the system, including automatic birthing, fetal ultrasound, fetal distress monitor, vital signs, Partographs, and heart and lung sounds.

[0070] Referring to FIG. 13, if one of the modules (FIG. 12) is selected by the user, such as by voice recognition or selection with a mouse of the computer 15, the program 15 a displays a display screen 416. The display screen 416 contains an information box 418, which contains topical information. The display screen 416 also has a menu bar 420 containing information items (illustrated as A-D for convenience) listing information categories specific to the topic of the selected module. It is understood that an item may be selected from the screen 416 via the menu bar 420, and that each module 414 a-p has its own display screen with its own menu of specific informational items A-D, which may be expanded to include a large number of items, or condensed for example, by placing selectable sub-items under an item.

[0071] Selection of an item from a menu, other than an exit item, causes text and/or illustrations topical to the selected menu item to be displayed in the information box 418. In practice, the program may generate a new display screen (not depicted). As such, it is understood that the information screen 416 is used as an example of any number of screens, and furthermore, such screens can be displayed in sequential order, or a series, for each item. A series of screens, such as screen 416, comprises a tutorial regarding patient treatment protocols for the selected menu item. Thus, the user can review information from a library of topics by selecting the appropriate module, and item, and then navigating through a series. Navigation in a series of screens is attained by the user's selection between three boxes: 422, 424, and 426, respectively “Back”, “Next”, and “Exit”, with corresponding function among the screens, such as proceeding backwards or forwards in the series. If no “Back” or “Next” function is possible, as respectively would be the case of the first and last screen of a series, the boxes 422 or 424 may be unselectable.

[0072] For example, modules 414 f and 414 g, each engender a series to teach a user about maternal and neonatal resuscitation, respectively. The user may also practice CPR on the simulator 14 (FIG. 1a), such as the maternal simulator 300, or the neonatal simulator 302′, above, and the program 15 a senses the user's compression and ventilation, via the CIM 16 (FIG. 1a) and sensors 30 (FIG. 1a). The heart and lungs of the simulator 14 are connected to pressure transducers confirming airway ventilation and cardiac compression; for example, an air line may be mounted in tracheal wall of the simulator 14 and connected to a sensor 30 connected to the CIM 16, so that when CPR ventilation is performed on the simulator, the CIM 16 monitors the timing and magnitude of the pressure and volume of the ventilation activity, via the air line and the sensor. Similarly, a compression bladder may be embedded within the chest cavity of the simulator 14 for sensing and confirming proper timing and magnitude of a CPR chest compression procedure, when connected by an air line to a compression sensor 30 attached to the CIM 16. The program 15 a compares the information pertaining to the user's activity with predetermined standards, and thus provides an interactive training session.

[0073] The predetermined standards are selectable, and reflect medical protocols used around the world, including BLS and ACLS guidelines set forth by the American Heart Association and others. At least seven major protocols for cardiopulmonary resuscitation (CPR) are stored and selectable by the user. Moreover, a user may update the protocols, or enter and store a “New Protocol” reflecting the local protocol regarding depth, duration, and frequency of cardiac compressions and airway ventilations. The program will use this series of acceptable limits to generate a new CPR waveform for testing CPR.

[0074] Referring back to FIG. 12, selection of a test module 414 h-j from the test box 410 directs execution of the program 15 a to provide a testing sequence to help test the user on patient care protocols, such as maternal and neonatal resuscitation, and other responses to emergency scenarios. The program 15 a paces through the steps of a patient distress scenario, giving the user a predetermined time to respond or complete the task required, thus enabling the user to experience the pressure of a emergency situation. For example, the program 15 a may test the user by presenting choices from which the user must select in order to treat the patient, wherein the user must complete the correct choice before the sequence proceeds to the next event. The program 15 a enables the user to enable, disable, or check the virtual instruments 12 and sensors 30 for connection to supply input to the CIM 16.

[0075] If the virtual instruments 12 (FIG. 2) are enabled, the user may implement patient care activity on the simulator 14 using the virtual instruments 12, while having the results and quality of response being monitored by the program 15 a. Alternatively, the user may use software-simulated instruments 12′ (FIG. 1b) generated by the program 15 a. The program 15 a advances through the scenario until the patient recovers, and provides a running critique of the user's responses, with an explanation of each incorrect choice or action. Features of the test modules 414 h-j include items that enable the user to specify that action sequences prescribed by the scenario comprise a predetermined number of compression/ventilation cycles on the simulator 14, or to allow the user to record the time and magnitude of the compression and ventilation activity performed on the simulator 14, or to select among a group of choices for hearing realistic sounds.

[0076] Testing may be defined by the program 15 a, as above, or by the user. For example, selection of the Codemaker Test module 414 j (FIG. 12) allows a first user, for example, an instructor, to create a scenario to test a second user, for example, a student. The first user may input preliminary data to define the patient simulator of the testing scenario by entering a set of preliminary patient parameters regarding information such as sex, weight, and age, as well as patient indications, vital signs and cardiac rhythms which will be realistically reflected in the vital signs monitor 406 (FIG. 12). An instructor defined testing system allows the instructor to test the student on local, national, or international patient care protocols. Many algorithms are selectable by opening files, including BLS, ACLS, Pediatric, and Obstetric (OB) emergencies. Other algorithms may be created and stored, and algorithms may be linked together as well. Benefits of this module include flexibility for instruction and the ability to detect mastery of the subject. An instructor-defined algorithm would presumably vary from well-known, structured algorithms, and thus avoid the problem of rote memorization of responses by the student.

[0077] Action may be taken in response to the conditions by the student, for example, the student may select among virtual instruments to use to render patient care activities. The student may then perform the patient care activities virtually, or using the tangible simulator.

[0078] Use of the modules 414 k-p of the virtual instruments tutor box 52 provides information about instruments commonly used in child birthing scenarios. In some instances, opportunities to practice using some of the virtual instruments 12 in patient care protocols with the simulator 14 are provided.

[0079] Turning now to FIGS. 14 and 15, the entire child birthing process may be automated via the program 15 a, with the user merely defining initial conditions, such as delivery time 430, delivery profile 432, and contraction intensity 434. The warp feature allows a full delivery to be condensed from 16 hours to 5 minutes. Child birthing then consists of placing the fetal simulator 302 on the projection 344, and placing the cover 324 on the maternal simulator 300. The program 15 a also offers a varying rate for progress of the ram 346, i.e., the first few centimeters may proceed much more slowly than the last few centimeters to better simulate child birth.

[0080] Referring to FIG. 16, if module 414 m (FIG. 12) is selected, a series of screens are shown regarding the fetal distress monitor, with tutorial information. An exemplary fetal distress monitor box 436 is depicted, along with a selectable On button 436 a for turning on the monitor. The fetal distress monitor 12 l cooperates with the simulator 14, the fetal heart monitor is placed on the cover 324 of the maternal simulator 300 (FIG. 5a) and interacts with at least one sensor 30, while the contractions monitor interacts with another sensor 30 disposed on the cover.

[0081] Although illustrative embodiments have been shown and described, a wide range of modification, change, and substitution is contemplated in the foregoing disclosure and in some instances, some features of the present embodiment may be employed without a corresponding use of the other features. It is understood that several variations may be made in the foregoing without departing from the scope of the embodiment. For example, the system 10 may be modified by simply modifying the program 15 a and/or the virtual instruments 30 and sensors 30. Accordingly, it is appropriate that the appended claims be construed broadly and in a manner consistent with the scope of the embodiment. 

What is claimed is:
 1. An interactive education system for teaching patient care to a user, the system comprising: a maternal simulator; a fetal simulator designed to be used both in conjunction with the maternal simulator and separate from the maternal simulator; and a neonatal simulator designed to replace the fetal simulator in post-birth simulations.
 2. The system of claim 1 further comprising a processor operable to execute a plurality of instructions associated with at least one of the maternal, fetal, and neonatal simulators, wherein the instructions include at least one scenario designed to instruct the user in a patient care activity.
 3. The system of claim 2 wherein the scenario is designed to utilize the maternal simulator to provide adult critical care training to the user.
 4. The system of claim 2 wherein the scenario is designed to utilize the maternal simulator to provide obstetrical training to the user.
 5. The system of claim 4 wherein the obstetrical training includes a plurality of obstetrical exercises, wherein the exercises include a breach exercise, a uterine palpitation and massage exercise, a vacuum extraction exercise, and a forceps delivery exercise.
 6. The system of claim 2 wherein the scenario is designed to utilize the fetal simulator to provide critical care training to the user.
 7. The system of claim 6 wherein the scenario is further designed to provide obstetrical training to the user when the fetal simulator is used in conjunction with the maternal simulator.
 8. The system of claim 2 wherein the scenario is designed to utilize the neonatal simulator to provide neonatal critical care training to the user.
 9. An interactive education system for teaching patient care to a user, the system comprising: at least one adult simulator for providing adult critical care and obstetrics training to the user; a fetal simulator for providing fetal critical care training to the user, wherein the fetal simulator can be used in conjunction with the adult simulator; and a neonatal simulator for providing neonatal critical care training to the user, wherein the neonatal simulator can be used in conjunction with the adult and fetal simulators.
 10. The system of claim 9 further comprising: a processor associated with at least one of the maternal and neonatal simulators, wherein the processor is operable to execute instructions corresponding to at least one patient care scenario; and an output means in communication with the processor, wherein the output means is operable to provide output based on the scenario and the user's interaction with the maternal simulator or the neonatal simulator.
 11. The system of claim 10 further comprising instructions for creating a custom scenario, wherein the instructions comprise: providing a graphical user interface having a plurality of menus, wherein each of the menus includes multiple options; enabling a scenario designer to select at least one option from the plurality of menus; and executing the scenario using the selected option.
 12. The system of claim 11 wherein the selected option is associated with a delivery profile for use in a birthing simulation.
 13. A method for teaching patient care to a user utilizing a maternal simulator, a fetal simulator, and a neonatal simulator, the method comprising: providing a plurality of adult care scenarios using the maternal simulator, wherein each adult care scenario presents the user with at least one adult critical care exercise or obstetrics exercise; providing a plurality of fetal care scenarios using the fetal simulator, wherein each fetal care scenario presents the user with at least one fetal critical care exercise; and providing a plurality of neonatal care scenarios using the neonatal simulator, wherein each neonatal care scenario presents the user with at least one neonatal critical care exercise.
 14. The method of claim 13 wherein the obstetrics exercise includes utilizing the fetal simulator in conjunction with the maternal simulator.
 15. A method for teaching obstetrics care to a user utilizing a maternal simulator, the method comprising: providing a plurality of training scenarios to a computer operatively coupled to the maternal simulator, wherein the training scenarios include a breach scenario, a uterine palpitation and massage scenario, a vacuum extraction scenario, and a forceps delivery scenario; selecting at least one of the plurality of training scenarios; and monitoring feedback provided by the maternal simulator to determine a quality of care achieved by the user with respect to the selected training scenario.
 16. The method of claim 15 further comprising selecting parameters for at least one of the plurality of training scenarios from a menu.
 17. The method of claim 16 wherein the menu is a pull down menu in a computer generated graphical user interface.
 18. The method of claim 15 further comprising designing a custom training scenario using a computer generated user interface.
 19. The method of claim 18 wherein the user interface utilizes a plurality of pull down menus having selectable options.
 20. The method of claim 19 further comprising enabling a scenario designer to select at least one option from the plurality of pull down menus, wherein execution of the custom training scenario includes the selected option.
 21. An interactive system for providing critical care and obstetrics training to a user using a plurality of patient care modules, the system comprising: a maternal simulator and a neonatal simulator, wherein the neonatal simulator can be used in conjunction with the maternal simulator; a processor operably connected to at least one of the maternal and neonatal simulators; and a memory accessible to the processor for storing instructions for execution by the processor, the instructions for: generating a user interface for displaying at least one of the plurality of patient care modules; enabling the user to select the displayed module; and executing the selected module.
 22. The interactive system of claim 21 further comprising instructions for providing a plurality of configuration options, wherein the configuration options enable the user to configure the selected module.
 23. The interactive system of claim 22 wherein the selected module is for obstetrics training, and wherein the configuration options include a delivery warp control configuration option, a delivery profile control configuration option, and a labor control configuration option.
 24. The interactive system of claim 23 wherein the delivery warp control configuration option enables the user to compress a delivery cycle into a specified time frame.
 25. The interactive system of claim 23 wherein the labor control configuration option comprises a frequency control, a duration control, and an intensity control.
 26. The interactive system of claim 25 wherein the labor control configuration option is divided into a plurality of phases, wherein the phases include a latent phase, an active phase, a transition phase, and a phase 2 phase.
 27. The interactive system of claim 22 further comprising instructions for implementing the configuration options using pull down menus.
 28. The interactive system of claim 22 wherein the configuration options enable the user to configure patient information and a treatment history.
 29. The interactive system of claim 28 wherein the patient information includes an age, a gender, and a weight.
 30. The interactive system of claim 28 wherein the treatment history includes a medication type.
 31. The interactive of claim 21 further comprising instructions for assigning each of the plurality of modules to a teaching group or a testing group.
 32. The interactive system of claim 31 wherein the teaching group includes a physiology module, a pregnancy module, a complications module, a labor and birth module, a postpartum module, a maternal resuscitation module, and a neonatal resuscitation module.
 33. The interactive system of claim 31 wherein the testing group includes a maternal resuscitation module, a neonatal resuscitation module, and a customizable module.
 34. The interactive system of claim 21 further comprising instructions for a plurality of tutorials, wherein the tutorials include an automatic birthing tutorial, a fetal ultrasound tutorial, a fetal distress monitor tutorial, a vital signs tutorial, a birthing records tutorial, and a heart/lung sound tutorial.
 35. The interactive system of claim 21 further comprising instructions for providing a plurality of sequential screen displays to the user and enabling the user to sequentially view the screen displays.
 36. The interactive system of claim 21 further comprising instructions for providing a menu having an instrumentation option, a log option, an ECG sounds option, a communications port option, a help option, and an exit option.
 37. The interactive system of claim 21 further comprising a fetal simulator, wherein the fetal simulator can be used in conjunction with the maternal simulator or separate from the maternal simulator.
 38. The interactive system of claim 21 further comprising instructions for providing feedback based on an action of the user with respect to the selected module and the associated maternal or neonatal simulator.
 39. The interactive system of claim 38 further comprising instructions for providing the feedback through a display device or an audio device.
 40. The interactive system of claim 39 further comprising instructions for providing feedback each time the user performs a predefined action, wherein the feedback informs the user of the action's success or failure.
 41. The interactive system of claim 40 further comprising instructions for detailing a rationale behind the action's success or failure.
 42. The interactive system of claim 40 wherein the action is associated with a local, a national, or an international patient care protocol.
 43. The interactive system of claim 39 further comprising instructions for presenting a question to the user, wherein the question is directed to training the user via the selected module.
 44. The interactive system of claim 39 wherein the display device displays at least one of an electrocardiograph sweep and a heart rate indicator.
 45. The interactive system of claim 39 further comprising instructions for displaying the use of an ultra sound transducer with the maternal simulator.
 46. The interactive system of claim 39 further comprising instructions for displaying a fetal heart rate, wherein the fetal heart rate is associated with a fetal simulator used in conjunction with the maternal unit.
 47. The interactive system of claim 39 further comprising instructions for a virtual instrument, wherein a position of the virtual instrument relative to at least one of the simulators results in a sound audible to the user.
 48. The interactive system of claim 47 wherein the sound replicates a physiological sound.
 49. The interactive system of claim 21 further comprising instructions for: detecting a sound produced by the user; and selecting the displayed module based on the sound.
 50. The interactive system of claim 49 further comprising instructions for enabling interaction between the user and the module based on a plurality of sounds produced by the user. 